Read or Download Agilent 35670A Analyzer (service) PDF
Best nonfiction_6 books
- Anal. of the Advantages of Changing Speed, Course (Submarines) [website capture]
- The Ruler in Geometrical Constructions (Popular Lectures in Mathematics)
- 360 SYSTEMS DigiCart II+ Digital Audio Hard Disk Recorder (DR-2750)
- Fire In The Blood (Harlequin Blaze)
- Stochastic Modeling and Control
Extra info for Agilent 35670A Analyzer (service)
Caution Do not apply any water mixture directly to the screen or allow moisture to go behind the front panel. Moisture behind the front panel will severely damage the instrument. To prevent damage to the screen, do not use cleaning solutions other than the above. To store the analyzer • Store the analyzer in a clean, dry, and static free environment. ’’ 2-17 Preparing the Analyzer for Use To transport the analyzer Agilent 35670A To transport the analyzer • Package the analyzer using the original factory packaging or packaging identical to the factory packaging.
Press the following keys to enable external mode: [ Disp Format ] [ MORE ] [ MORE ] [ EXT MON ON OFF ] Pin Number Signal Name Pin Number Signal Name 3 R 8 HSYNC 4 G 9 VSYNC 5 B 1, 2, 6 GND † The EZPIXpc driver converts TTL video signals into RGB analog signals, drives 75 ohm coax cable, provides RGB composite sync or RGB sync on green, for monitors with RGB input capability. , 1725 West 17th St, Tempe, Arizona 85281, 800-638-6104 2-13 Preparing the Analyzer for Use To connect the optional keyboard Agilent 35670A To connect the optional keyboard The analyzer may be connected to an optional external keyboard.
Using the dc power cable (HP 35250A), attach the black cable to the common terminal and the red cable to the positive terminal of the dc power source. Using the dc power cable with cigarette lighter adapter (HP 35251A), plug the cigarette lighter adapter into an automotive cigarette lighter receptacle. • Connect the analyzer’s ground terminal to the same reference potential as the common terminal of the dc power source. Using a wire, connect the analyzer’s GROUND terminal to the common terminal of the dc source.